This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3. This manual provides documentation for the following instruments: PSA Series. EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3.
|Published (Last):||1 April 2008|
|PDF File Size:||14.36 Mb|
|ePub File Size:||10.71 Mb|
|Price:||Free* [*Free Regsitration Required]|
Also, listen for internal fan noise to determine if the analyzer cooling fans are running. Page 25 Item Description Data invalid indicator Status message area Key menu title Softkey menu Frequency span or stop frequency Sweep time Video bandwidth Frequency offset Display status line Resolution bandwidth Center frequency or start frequency Internal preamp Auto alignment routine status Trigger or Trace Average Display line annotation This symbol is used manul indicate the ON position of the power button.
Put a formatted floppy in the A: Front and Rear Panel Features This chapter gives you an overview of the front and rear panels of your analyzer. It is effective over a frequency range of dc to 2 GHz.
Factory Preset Restores the analyzer to a user-defined state. To access the menu, press Rename 2. By pressing ‘print’ button you will print only current page. The battery is located on the CPU board. Line voltage does not need to be selected. Retain the shipping materials for future use, as you may wish to ship the analyzer to another location or to Agilent Technologies for e444a0.
Select the type of file you want to save. In this example you will save a trace to the internal C: Front and Rear Panel Features Key Overview In other key menus, one key label will always be highlighted to show which key has been selected but the menu is immediately exited when a selection is made. Additional Information Notice The information contained in this document is subject to change without notice.
Please be as specific as possible about maunal problem.
Agilent Technologies E4440A Getting Started Manual
The mains plug shall be inserted only in a socket outlet provided with a protective earth contact. Index personality options not in instrument power amplifiers preamplifiers preset factory types user, creating printer setup probes AC and DC product information rear panel symbols rear panel features reference level, setting reference, turning on 10 MHz rename files RF limiters Press If the previously performed alignments did not resolve the problem, press Alignments Restore Align Defaults Is the analyzer displaying an error message?
Protecting Against Electrostatic Discharge Protecting Against Electrostatic Discharge Electrostatic discharge ESD can damage or destroy electronic components the possibility of unseen damage caused by ESD is present whenever components are transported, stored, or used.
Select the type of file you want to rename setup, state, trace, limits, screens, or corrections. To access the Load menu, press 3. If you wish to verify this condition, remove the signal e4440aa.
Agilent Technologies makes no warranty of any kind with regard to this material, including but not limited to, the implied warranties of merchantability and fitness for a particular purpose.
E4440A PSA Spectrum Analyzer, 3 Hz to 26.5 GHz
Turning on the Analyzer for the First Time Plug in the power cord. Set the frequency span to 50 MHz: In this example you are loading a trace file; Page 30 Front and Rear Panel Features Key Overview In other key janual, one key label will always be highlighted to show e444a key has been selected but the menu is immediately exited when a selection is made.
Place a marker labeled 1 on the 10 MHz peak, as shown in Press Peak Search Note that the frequency and amplitude of the marker appear both in the active function block, and in the upper-right corner of the screen.
Accessories Sales and Service Office and are listed below.
Used Agilent EA
Used on Option Noise Figure personality. Creating A Directory b. Viewing Traces 1 and 2 7. Types of Spectrum Analyzer Messages The analyzer can generate various messages that appear on the display during operation.
Page 73 Perform an Alignment. This symbol is used to indicate the OFF position of the power button. This manual provides documentation for the following instruments:. They are only designated by manuao Name: Like PCs, these analyzers include an internal storage drive and a floppy disk drive, both of which have directory and sub-directory capability.
If your analyzer is covered by a separate maintenance agreement, please be familiar with its terms.
AGILENT TECHNOLOGIES EA GETTING STARTED MANUAL Pdf Download.
Comments to this Manuals Your Name. Firmware Revision Firmware Revision To view the firmware revision of your analyzer, press call Agilent Technologies regarding your analyzer, it is helpful to have this revision and the analyzer serial number available. If the necessary equipment is available, perform the functional tests in the Instrument Messages and Functional Tests document for your analyzer. Page e4440w When you have completed the printer setup for the analyzer, press the front panel key.
Available only at time of purchase Chapter 5 Description Provides the factory calibration test data and the standard commercial calibration certificate at initial shipment. Directories are displayed as DIR and drives have nothing listed under Type:.
Page 23 Item Name Access features used with all analyzer modes and affect the state of the entire spectrum System Keys analyzer. Page 70 In Case of Difficulty Before calling Agilent Technologies, however or returning an analyzer for serviceperform the quick checks listed in the problem. Sales and Service Office and are listed below. For example, when you press the key on the menuit will bring up its own menu of Orientation To access the Copy menu, press boxes as described by the From: Select the type of file you want to load setup, state, trace, limits, or corrections.
Each softkey More has several characters on it. Making a Basic Measurement This chapter is divided into the following sections: Options And Accessories Options and Accessories This chapter lists options and accessories available for your analyzer.